PROFIBUS Protocol Converter
IMPAC protocol converter IMPAC protocol (RS232 or RS485) <-> Profibus DP for 1 pyrometer only
Reference number: 3 852 440
The RS485 – PROFIBUS DP – adapter is designed for the connection a digital pyrometer made by company LUMASENSE to the PROFIBUS DP network. Devices equipped with RS232 and RS485 UPP interfaces can be connected to the adapter.
The pyrometer can be read and parameters can be passed to it via the PROFIBUS. The adapter operates as master on the RS485 bus and as slave on the PROFIBUS network. The data is transferred between the pyrometer and the adapter via an RS485 or RS232 interface.
The data from the adapter to the PLC is transferred using the PROFIBUS protocol.
The length of the cable between the pyrometer and the adapter should not exceed 20m when devices with an RS232 interface are used.
The pyrometer must be operated at the factory default of 19200 Baud and address 00.