IMPAC protocol converter IMPAC protocol (RS232 or RS485) <-> ProfiNet for 1 pyrometer only
Reference number: 3 852 620
The RS485 – PROFINET IO – adapter is designed for the connection a digital INFRATHERM pyrometer made by company LUMASENSE to the PROFINET network.
Devices equipped with RS232 and RS485 UPP interfaces can be connected to the adapter. The pyrometer can be read and parameters can be passed to it via the PROFINET. The adapter operates as master on the RS485 bus and as slave on the PROFINET network. The data is transferred between the pyrometer and the adapter via an RS485 or RS232 interface.
The data from the adapter to the PLC is transferred using the PROFINET protocol. The device features 2 Profinet-ports - one port can for instance be used as outgoing Profinet-port.
The length of the cable between the pyrometer and the adapter should not exceed 20m when devices with an RS232 interface are used.
The pyrometer must be operated at the factory default of 19200 Baud and address 00.